DUAL CHANNEL, ANALOG TENS
WITH 3 MODES, ECONOMY-PRICED
Dual-channel, 3-mode Analog TENS
Analog dials, adjustable pulse width, pulse rate, and timer, with sliding case and channel intensity safety cover.
Includes 2 lead wires and 4 electrodes.
FDA-listed, with Canadian Medical Device License #87575
The TENS 3000 is an analog TENS (transcutaneous electrical nerve stimulation) unit with the easiness of dial controls, yet with the accuracy of a digital unit. Adjustable pulse rate, pulse width, timer, and 3 modes all help you control stimulation for the very best results to relieve your particular type of pain.
The TENS 3000 features a sliding case design, and comes in a hard carry case complete with batteries and accessories.
A prescription is required to buy TENS units in the U.S. Use our Prescription Form, or have your health care provider supply a form.
9V Alkaline Battery
TENS 3000 stimulator
2 Lead wires (45”)
4 adhesive electrode pads (pack of 4)
Channel 1 Output Jack
Intensity Control Cover
Channel 2 Output Jack
Intensity Control Dials/ Power Switch for Channels 1 and 2
Indicator Lights for Channels 1 & 2
Mode Selector Switch
Pulse Rate Control
Pulse Width Control
9V Battery (brand varies)
Timer / Continuous Switch
Sliding Case Cover
Channel Intensity Control Dials (D) adjustable from OFF to level 8.
Mode Switch (F) allows selection of Burst, Constant, and Modulation modes.
Pulse Width Control (H) allows adjusting from 30-260 microseconds
Pulse Rate Control (G) al- lows adjusting from 2-150Hz
Timer Switch (J) allows selection of 15 or 30-min timer, or continuous stimulation.
SPECIFICATIONS FOR THE TENS 3000
2, isolated between channels
3 TENS modes (Burst, Constant, and Modutation)
Adjustable, 0-80 mA peak to 500 ohm load / channel
Adjustable from 2-150 Hz
From 30-260 µs
Asymmetrical bi-phasic square pulse
9 V Battery, included
0-40 V / load 500 ohm
CHARGE PER PULSE
Maximum 20 microcoulombs
(H X W X D)
3.7 X 2.5 X 1" /
95 X 65 X 24 mm
115g with battery
Selectable between 15 or 30 minutes, or continuous
BURST MODE (B)
Bursts occur twice every second, pulse width adjustable, frequency fixed at 100Hz
NORMAL MODE (N)
Pulse rate is automatically varied in 10-second looping cycle as follows: pulse rate decreases linearly by 40% over 4 seconds, then remains constant for 1 second before increasing linearly back to its original value over a 4-second period.